Designed with your safety in mind. Protecting the operator’s body, hand and other areas close to the device radiation exposure with our double shield protection.
Works with all plates and sensors
Four buttons to memorize your needs. Set up the exposure values onto each button. If you have a typical patient type, categorize them into a specific button Adult/Child, and their type of teeth; upper or lower section of the mouth, Canine tooth or molar for another.
Compact & Light
Under 6lbs Lighter then most handheld portable generators in the market.
Extremely mobile wireless handheld unit.
(optional mobile stand available)
Highest Level of Safety
Designed with yours and the patients safety in mind. Safety is ensured by internal double shielding, ensuring operators safety from the leakage radiation.
Aid to Diagnose
Clear X-Ray image is obtained while decreasing the radiation dose.
Large Battery Capacity
More than 300 exposures in one battery charge.
400,000 exposures a year equals 20 mSv. The operator dose limit per year is 50mSv. Makingyour safety our priority. One time exposure is 0.00005mSV
Slim Retraction Head
Registration Number K132041
Approved by international quality organization such as CE, KFDA, and FDA.
Focal spot size is an important factor to get a clear image in any X-Ray. By using Toshiba D-041 tube, unit provides smaller focal spot (0.4 mm) and smaller divergence of X-Ray (12°). You will experience the most sharp and clear images.
Smaller focal spot with the smaller angle degree can provide the sharpest images whatever could be. Because smaller divergence angle of the X-ray beams make edges of the image on the detector more accurately. You can get higher contrast and density of dental X-ray image by using the focal spot 0.4 mm comparing with 0.8mm portable X-ray devices.
Tube voltage is the power of penetrating the bone. In the case of cheek bone, 70 kV could be the reason to cause unclear images as the tube power is not enough to penetrate.
Tube voltage is the power of penetrating the bone. Powerful voltage helps you obtain high-resolution X-ray image. The main advantage of powerful voltage is to eliminate blurred and unclear pictures.
Current is a flow of electric charge which means the quantity. In order to get a clear image, the power and current of the tube is and important factor.
Current is a flow of electric charge which means the quantity. In order to get a clear image, current of tube is also important factor. Moderate quantity of X-ray beams helps to capture the image more clearly.
Don’t Worry, We are Compatible
With all digital USB type sensors as well as conventional film and phosphor plates.
The Polaroid Rextar Portable Handheld X-Ray Generator:
Generator, Back Scatter Safety Shield, Long Cone, Neck-belt,
Hand Strap, Carrying Bag, Battery Charger, Hand Switch
(Optional Remote Controller), and User Manual.
Weight: 2 kg (Main Body+Built-in Battery)
Back-Scatter Shield: 12mm Thick, x 6″ Diameter
Short Cone: 40mm
Long Cone: 140mm
Output Specification: 140 W
Device Power Source: 11.1 VDC (Battery)
Battery: Lithium-Ion Polymer
X-Ray Exposure Time Range: 0.01 ~ 1.30 sec ( 43 Step )
Display: LCD Panel Display
Tube Inherent Filtration: 1.0mmAL Eq.
Addition Filtration: 0.5mmAL
Total : 1.5mm AL Eq. @ 70 kV
Device Power Source: 19 VDC 3.16A
Charger Power: 100-240 VAC, 50-60Hz 1.5A
kV: ± 7%
Sec: ± 10%
Name: Toshiba D – 041
Tube Voltage: ±4% KV
Tube Current: ±10% mA
Tube kV/Ma= 70kV/2mA(fixed)
Voltage Range (Maximum Voltage): 50 ~ 70kV ( 77kV )
Focal Spot Size: 0.4 × 0.4mm
Input Electrical Power (Per Second) : 430W
Inherent Filtration: Min. 1.0mm AL Eq.
Anode Heat Radiation Index: 4300J
Maximum Rate of Anode Heat Exhaustion : 100W
Warranty = 2-Yr Manufacturer’s Defect Warranty
Remote Support = Free Lifetime Support
Looking for more X-Ray equipment?
Click below toshop our collection!||check out our newest tech!||discover what Polaroid Healthcare has to offer!
Portable X-Ray Generator
We're here to help.
Get in touch with our customer service team.